The HL-2000-CAL Calibrated Tungsten Halogen Light Source can calibrate the absolute spectral response of a radiometric system. With this NIST-traceable light source and SpectraSuite, you can determine known absolute intensity values at wavelengths from 300-1050 nm. The HL-2000-CAL Calibrated Tungsten Halogen Light Source can be used to calibrate the absolute spectral response of a radiometric system. With this NIST-traceable light source and SpectraSuite Software, you can determine known absolute intensity values at wavelengths from 300-1050 nm.
NIST-traceable Calibration from 300-1050 nm. Provides absolute spectral intensity in µW/cm2/nm at the fiber port
Source can be calibrated specifically for a bare fiber, a fiber plus cosine corrector, or an integrating sphere
Calibration Certificate. Calibration data is provided in paper and electronic formats, for use with our our OOIIrrad-C Irradiance Software or SpectraSuite
Calibrated for Use with Fiber and Cosine Corrector
The HL-2000-CAL is designed for calibrating a system consisting of a spectrometer and a cosine corrector. The calibration data for the HL-2000-CAL includes absolute intensities for wavelengths between 300-1050 nm.
Calibrated for Use with Integrating Spheres
The HL-2000-CAL-ISP is designed for calibrating the absolute spectral response of your system when using the ISP-50-8-I Integrating Sphere as your sampling optic.
Included with the Lamp
The HL-2000-CAL comes with a regulated 12 VDC power supply. Also included is a calibration certificate and a diskette with data files for use with our irradiance software. A cosine corrector is available for an additional charge .
The typical calibration output when using the HL-2000-CAL with a CC-3 Cosine Corrector.
|Spectral range (calibrated):||300-1050 nm (calibrated)|
600 mA @ 12 VDC
|Power output:||6.5 watts|
900 hours (recommend recalibration after 50 hours of use)
|Recalibration:||Required after ~50 hours of operation|
|Bulb color temperature:||
|Output to bulb:||5 volts/1.3 amps|
|Output regulation:||0.2% voltage|
|Time to stabilized output:||~30 minutes|
|Connector:||SMA 905 for fiber; 6.35-mm barrel for cosine corrector; PTFE plug for integrating sphere|